Megger DLRO600 measures resistances between 0.1 µO and 1 O, at high currents of up to 600 amps.
This versatile instrument can provide test currents from 10 amps up to 600 amps subject to the load resistance and supply voltage.
A large liquid crystal display provides all the information needed to perform a test; all test parameters and measurement results are displayed. The unique design allows the weight and size of DLRO600 to be kept to a minimum; the instrument weighs less than 15kg. This small size makes DLRO600 equally at home in the workshop, on the production floor or in the field. The high current capability and compact design make DLRO600 suitable for testing circuit breaker contacts, switch contacts, busbar joints or other applications where high current is needed.
300 sets of results may be stored in DLRO600’s on board memory for later download to a PC or may be output directly to a printer via the RS232 port. You may also add notes to any stored result by using the on board alphanumeric keypad, thereby making later identification of results straightforward.
As well as adding notes to stored results, the alphanumeric keypad allows you to set the test current directly by typing in the value required. DLRO600 will check the continuity of the test circuit, and will quickly ramp the test current up to the desired level. The keyboard is also used to set upper and lower limits for the result and
to prevent the use of excessive currents by setting an upper limit to the allowable test current.
DLRO600 uses a four terminal measurement technique to cancel the resistance of the test leads from the measurement.
DLRO600 operates in one of three modes, which are simply selected from the on screen menu.
CONTINUOUS mode is provided for users who wish to monitor a resistance over a period of time. Connect the test leads, select the test current and press the TEST button. DLRO600 will pass a current continuously, and measure the resulting voltage at 2 second intervals, until the test button is pressed to stop the test or the test
circuit is interrupted.
In NORMAL mode you connect the leads, select the test current and press the TEST button. The test current will ramp up to the desired level, hold for 2 seconds and then ramp down. The whole process takes approximately 7 seconds.
In AUTO mode select the desired current, connect the current leads and press the TEST button. The TEST lamp will flash to show that the DLRO600 is ready to carry out a test. As soon as the potential leads are connected, a test will start. To repeat a test, simply break contact with the voltage probes and remake contact.
Measuring individual joints in a busbar is a good example of the convenience to be gained by using AUTO mode. The two current leads are connected to the ends of the busbar. They will remain connected here until all tests have been completed. When the voltage leads make contact across a joint, DLRO600 detects that all four leads are connected, carries out a test and stops. When you move to the next joint DLRO detects the new completed circuit automatically and carries out the next test, and so on until all joints have been tested. The results may be stored automatically and may be recalled to the display or downloaded for review.
- Test currents from 10A to the highest 600A test current
- 0.1µΩ best resolution
- Range of 0.1µΩ to 999.9mΩ
- Store up to 300 test results and notes onthe DLRO600's onboard memory
- Alpha-numeric keypad facilitates easy entry of test notes
- RS232 port to download stored results to PC or real time output to a printer
- Three test modes, including an AUTO mode
- Easy to set test current by directly typing in via keypad
- Large, high resolution backlit LCD display
- Compact footprint and weighs less than 15kg
- Supplied complete with 5 metre test leads and download software